Reliable identification in GC-MS analysis with two independent analytical information sources
contributed by Shimadzu |
Main flavour and fragrance components prior to gas chromatographic (GC) separation are usually identified with mass spectrometry (MS) using spectral comparison and literature data. In the GC - MS peak identification of complex samples, the main difficulty is related to the almost identical mass spectra of terpenes. Hence, MS identification should be accompanied by additional information, namely retention index (RI), that can support the MS similarity search. There are several ways to calculate the RI of an unknown compound according to the specific chromatographic conditions [1-4]. The widely accepted way for programmed temperature retention index, denominated also as linear retention index (LRI), calculation is expressed as follows [4, 5]:
where tR is the retention time, z is carbon atom number, i is the peak of interest, n is the difference in carbon atom number of the two standards (n -alkanes : n =1, even or odd fatty acid methyl esters (FAMEs) and fatty acid ethyl esters (FAEEs) : n=2). The Flavour and Fragrance Natural and Synthetic Compounds Library (FFNSC) is a comprehensive database containing mass spectrum of about 3500 individual flavour and fragrance molecules, acquired mainly from pure standards. Moreover, for each compound embedded experimental LRI values on three different stationary phases (SLB- 5ms, Equity -1 and Supelcowax -10) are also reported. Boosting the MS similarity search with simultaneous LRI filter of GCMSsolution a fast and reliable peak assignment can be achieved. Using two different types of stationary phases and the MS spectra and exploiting the multi-LRI feature of the FFNSC Library the identification will not be tentative, but confirmed by three individual analytical information.
The FFNSC Library can be an excellent support and applied in essential oil characterization, food aroma profile analysis, determination of fragrances in non- food matrix (e-liquids, cosmetics). In this application note some examples are reported.
Enjoy our FREE content!
Log in or register to read this article in full and gain access to The Analytical Scientist’s entire content archive. It’s FREE and always will be!
Login if you already created an account
Or register now - it’s free and always will be!
You will benefit from:
- Unlimited access to ALL articles
- News, interviews & opinions from leading industry experts
- Receive print (and PDF) copies of The Analytical Scientist magazine